Trinocular LED Fluorescent Biological Microscope
Compensation Free Trinocular Head, Wide Field, Epi-Fluorescent Attachment and Infinite Semi-Plan Achromatic Objectives and Infinite Optical System
LB-245 are newly developed microscopes with an infinity color corrected system , the microscopes use LED as the light source, the life span of the LED lamp is much longer than mercury lamp, the performance is also better.
|View Brochure by Flipbook|
|PDF Brochure – Page 1||PDF Brochure – Page 2|
Labomed, Inc. is certified by ISO 9001-2013, has CE Conformity and is FDA Licensed.
Excellent fluorescent image with infinitive optical system.
LED illumination with 50,000 hours life span.
No need for cooling system and protection device due to LED cold light source.
465-476nm excited wavelength with high SNR.
Convenient to use, no need warm-up.
More fluorescent filters are available for different applications.
LB-245 Trinocular LED Fluorescent Biological Microscope with Compensation Free Trinocular Head, Wide Field, Epi-Fluorescent Attachment and Infinite Semi-Plan Achromatic Objectives and Infinite Optical System is used to study the absorbing, transportation, chemicals distribution and positioning in cells. It is widely used in disease examination, immune diagnosis and life science areas.
|Optical System||Infinite Optical System|
|Viewing Head||Compensation Free Binocular Head Inclined at 30°，Interpupillary Distance 48-75mm|
|Eyepiece||Wide Field Eyepiece WF10×/18|
|Nosepiece||Backward Quadruple Nosepiece|
|Objective||Infinite Semi-plan Achromatic Objective 4× 10× 40× 100×|
|Condenser||Condenser NA 1.25|
|Focusing||Coaxial coarse & fine adjustment, Fine division 0.002mm, Coarse Stroke 37.7mm per rotation, Fine Stroke 0.2mm per rotation, Moving Range 20mm|
|Stage||Double Layers Mechanical Stage 140×140/ 75×50 mm|
|Illumination||Halogen Lamp 6V/20W|
|Reflected Light Source||Excitation||Dichroic Mirror||Barrier Filter|
|Lamp||3W LED Lamp(465-476nm)|
|Immersion Oil||Fluorescent Free Oil|
ALL ACCESSORIES ARE OPTIONAL AND MUST BE ORDERED SEPARATELY, UNLESS OTHERWISE STATED IN THE TECHNICAL SPECIFICATIONS